- Project plans
- Project activities
- Legislation and standards
- Industry context
Last edited 16 Sep 2020
It consists of a hollow copper sphere painted matt black to absorb radiant heat, with a temperature sensor at its centre. When it reaches a steady state (after 15 minutes or so depending on the size of the globe and the environmental conditions) the heat exchanges by convection and radiation will be in equilibrium, and the temperature recorded by the sensor will be somewhere between the air and radiant temperature. This is referred to as the globe temperature (tg) or black globe temperature and resembles the thermal conditions felt by the human body.
MRT = tg + 2.42 x air velocity in m/s (globe temperature – air temperature)
(equation ref Novalynx Corporation 2010 - this equation is dependent on the size and emissivity of the globe thermometer, and the units they show for air velocity are incorrect - cm/s, rather than the correct m/s shown above)
 Find out more
 Related articles on Designing Buildings Wiki
Featured articles and news
Developments in the Future Homes Standard.
An American chimney feature with a colourful past.
Homes based on need, not ability to pay.
Historic England adds 216 entries to the 'at risk' register.
Will cycling and walking provisions be preserved?
Assembly point levels range from relative to ultimate.
Signs are pointing to a recovery for the construction industry.
Campaigning to change perceptions about American Brutalism.
Sprinkler head configurations can prioritise people or property.
Report from The Carbon Project reveals shortcomings and recommendations.
Advice on how to join the electrotechnical profession.
BREEAM Building Back Better briefing paper provides initial guidance.
CIOB conducts global search for best built environment picture.