- Project plans
- Project activities
- Legislation and standards
- Industry context
Last edited 12 May 2019
It consists of a hollow copper sphere painted matt black to absorb radiant heat, with a temperature sensor at its centre. When it reaches a steady state (after 15 minutes or so depending on the size of the globe and the environmental conditions) the heat exchanges by convection and radiation will be in equilibrium, and the temperature recorded by the sensor will be somewhere between the air and radiant temperature. This is referred to as the globe temperature (tg) or black globe temperature and resembles the thermal conditions felt by the human body.
MRT = tg + 2.42 x air velocity in m/s (globe temperature – air temperature)
(equation ref Novalynx Corporation 2010 - this equation is dependent on the size and emissivity of the globe thermometer, and the units they show for air velocity are incorrect - cm/s, rather than the correct m/s shown above)
 Find out more
 Related articles on Designing Buildings Wiki
Featured articles and news
Guidance for local authorities and consultancies setting planning conditions.
A real deal – at last?
How does anastylosis help in the reconstructing of ancient monuments?
More than just aesthetic and historic values and meanings.
An exciting and novel collaboration between the RIBA and the SPAB.
Republic of Ireland updates to planning and development.
The different types of pile foundation.
Achieving a net-zero carbon UK by 2050.
Responding to an invitation to tender.
Statutory instruments laid in Parliament to amend the Climate Change Act.
How will we pay for infrastructure post-Brexit after EIB has gone?